TS–013 Testing high capacity weighing instruments to maximum
To provide instruction on testing high capacity weighing instruments to the maximum capacity.
The capacity required to be reached when testing weighing instruments is detailed within Table 1 of this policy.
Where an instrument cannot be tested to maximum capacity, the reason must be recorded on the test sheet.
|Capacity of instrument||Minimum capacity to be tested to|
|< 40,000 kg||40,000 kg|
|> 40,000 kg||Within 10% of maximum capacity|
Trading Standards expects instruments to be tested to the approved maximum capacity. This policy is only to be used where there is insufficient substitution load available.
Trade Measurement — Technical Policy
|Policy number:||TS - 013|
|Effective date:||July 2016|
|Policy name:||Testing high capacity weighing instruments to maximum|
|Reviewed by:||Davis White, Legal Metrology Advisor and Senior Trading Standards Officer|
|Approved by:||Phil Sorrell, Team Leader Compliance and Inspection|